Sergei Bezrukov writes:
This is a shortened translation from Russian of my article published in Радио (Radio) magazine, no. 8 (2010) pp. 21-23. The voltmeter is designed for controlling DC voltages at the output of a dual bipolar power supply. It not a universal instrument, since its input resistance does not exceed 40K, which might be too low for some measurements. However, it is perfectly fine for power supplies and does not noticeably load their output. The range of input voltages is 0 to 24V for positive supply and -24V to 0 for negative one.
The voltages are sampled every 300 msec and displayed on a 9-digit LED display. Only 8 digits are used, the middle digit slot is always off and serves a a separator between negative (on the left part of display) and positive (on the right one) readings. The unit provides a 0.01V resolution. For small negative voltages (not exceeding 10V in absolute value) a leading minus sign is displayed. For positive voltages a leading zero is suppressed.
Voltmeter for bipolar power supplies - [Link]