In this episode Shahriar investigates a peculiar problem with a SRS DS345 function generator. The instrument does not appear to produce the correct output signal frequency. After some investigation it becomes clear that a few of the display digits are not active which hide the actual frequency settings. The problem is traced to a broken resistor on the display driver. The schematic of the instrument shows that the resistor is responsible for strobing the affected seven-segment digit. After the repair, the instrument’s performance is verified including the OCXO accuracy.
Teardown & Repair of a SRS DS345 30MHz Synthesized Function Generator – [Link]
Our friends at educ8s.tv uploaded a new tutorial on their youtube channel. It’s about an ESP32 web server along with MBE280 sensor.
Welcome to another ESP32 video tutorial! In this video, we are going to build a simple HTTP Web Server on an ESP32 board with a BME280 sensor. We are also going to learn how to make some requests to it using a web browser. There is a lot to cover, so let’s get started!
ESP32 Web Server Tutorial with a BME280 Sensor – [Link]
In this episode Shahriar & Rosanah investigate an Agilent power supply which does not appear to power on. It can be quickly observed that the fuse has failed on the unit. Using an isolation transformer a small amount of AC voltage is applied to the unit after the fuse replacement. It is clear that a short is present somewhere in the instrument since even at 10V AC the instrument consumes more than 1A.
Teardown & Repair of an Agilent E3632A DC Power Supply – [Link]
In this episode Shahriar repairs an Agilent PSA Series Spectrum Analyzer. The instrument generates many errors during self-alignment and produces no measurements below 3.2GHz. The block diagram of the unit is thoroughly presented and various possible failure points are considered. Based on the observation of the noise floor, the most likely cause is the second LO module. The measurement of the LO power indicates that the second LO power is fall below nominal.
Teardown, Repair & Analysis of an Agilent E4443A 3Hz – 6.7GHz PSA Series Spectrum Analyzer – [Link]
In this episode Shahriar investigates a peculiar problem with an Agilent P-Series Power Meter. While the instruments works during startup, after about 10 minutes the LCD screen begins to flicker and become scrambled. Initial investigation reveals that the issue is not likely with the main motherboard since USB connection to the instrument and data-readout is possible even when the LCD screen is malfunctioning.
Teardown & Repair of an Agilent N1912A P-Series Power Meter – [Link]